Micro moisture meter is a new development of the determination of trace water analysis instrument, the instrument uses high resolution color touch LCD display, convenient man-machine conversation, intuitive and easy to operate.The instrument adopts the high performance ARM processor with large data storage capacity, running fast and stable performance and excellent anti - jamming performance, which has outstanding advantages of high detection speed and high precision. The instrument has the fault diagnosis function, the end of the test, display and print results.
electric quantity titration (Coulomb analysis)
|100mg ~ 0ug (10ug ~ 100ug)|
100ug + 3UG, 500ug + 3% (no error, environmental humidity error)
Test sample type
solid, liquid, ga
64K color high definition touch monitor
1000 test record
Stir mixing speed
sliding touch panel speed control
ten years of normal operation of power down tim
micro thermal printer, paper width 56m
220V 10V + AC, 50Hz + 2.5H
|Using ambient temperature|
|Humidity||less than 85%|
260mm X 220mm X 330mm (length x width x height).
1.The modified transformer oil tester has a large-screen liquid crystal display, an interactive interface, and various parameters with menu prompts;
2The modified transformer oil tester test date with various parameters, clock and other tips;
3.differential detection, the system automatically corrects the deviation;
4.mixing, testing, computer automatically print data;
5.micrograms of water and ppm units, automatically converted;
6.the introduction of imported XECOM US data processing chip, fast and accurate data processing;
7.Instrument has potential is measured dynamic curve indicating function, makes the test more intuitive; instruments and data storage capacity up to store 500 data records; instrument has delay measuring function, is very effective in the testing of lower moisture content of specimen; instrument adopts the sliding touch control the stirring speed control; water content calculation formula contains calculated according to the volume and weight of the key parameters of various algorithms; testing process, such as the need to modify the formula for calculating the related parameters, can be modified in a timely manner and does not affect the measurement of moisture content, moisture content is in accordance with the revised parameters calculated to facilitate the users.